The Rheumatoid Factor is a high diagnostic interest biological marker for adult rheumatoid arthritis.
ELITech developed the ELI.H.A RF, an indirect hemagglutination assay, for the quantitative determination of IgM Rheumatoid Factor in serum. It is rapid, easy-to-use and easy-to-read with results available within 2 hours.
ELI.H.A RF was specially designed to eliminate interferences due to natural anti-sheep agglutinins.
ELI.H.A RF is based on the Rheumatoid Factor specific haemagglutination properties, used in reactions of the Waaler-Rose type.
Sheep red blood cells (Reagent 1) are sensitized by a gamma-globulins fraction of rabbit anti-sheep red blood cells serum.
The presence of Rheumatoid Factor in serum samples is revealed by a haemagglutination of the reagent R1: a reddish-brown film can be observed in the well. In the absence of Rheumatoid Factor, patients, these red blood cells deposit, forming a button in well bottom.
Non-sensitized red blood cells (reagent R2) are also available in the kit. They ensure the specificity of the reaction by allowing identifying any interference from the natural anti-sheep agglutinins (Forssman heteroantibodies, infectious mononucleosis antibodies…).
The reaction is carried out in a U-microplate.
Each kit allows 220 tests to be carried out or 20 reactions of 11 dilutions. Results are obtained within 2 hours and are reliable overnight.
If an agglutination is observed in well 11 (serum control), then remake a test with a preliminary adsorption of the natural anti-sheep agglutinins (Reagent 3).
Easy-to-Read and Easy-to-Interpret Result
Absence of haemagglutination.
Presence of a button at the bottom of the well.
Presence of haemagglutination.
Presence of a cloudy red/brown deposit coating the well.
Reagents and Material
R1: Vial of 4.4 mL of sensitized red blood cellsR2: Vial of 1 mL of non-sensitized red blood cells
BUF: Vial of 55 mL of phosphate buffer pH 7.2
R3: Vial of 2 mL of adsorbent
CONTROL +: Vial of 0.5 mL of titrated positive control
CONTROL –: Vial of 0.5 mL of negative control
MICROPLATE : Microplate with a U-bottom
DROPPER: Special dropper
Material Required but Not Supplied
Automatic pipette(s) with a pipetting volume adapted to the volume that will be measured
Contaminated waste containers
Stability and Storage
The reagents are ready-to-use.
All the reagents are stored at 2-8°C. In their original packaging, are stable until the expiry date indicated on the box.
There are no reviews yet.